Make sure to read chapter 1 “Installation” in the user manual after downloading and before installing by executing “setup. Measurement Speed and Range are left at their default values. The following photo shows the detector on a small plastic lens that is held in place by a piece of modeling clay plasticine. What industry are you in? Turn the unit upside-down, push the probe end of the detector in slightly to release it, and then lift the detector up. Second, attach the provided cable to the display unit and the detector.
|Date Added:||2 December 2018|
|File Size:||48.29 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Be careful not to touch the probe. Communication between the computer and the SJ is via a USB cable exactly like the ones commonly used for cameras.
Software and Updates
I hereby declare my consent that my data is used for dispatching information and advertisement on industrial measuring technology via e-mail by Mitutoyo Europe GmbH. Do you have a product enquiry? The most important thing is a way to hold the instrument so you can make measurements.
This is a usg option if you have a lot of similar parts to measure.
To achieve maximum usefulness, you will probably need to buy some more parts. Parameter may ,itutoyo left at 3 and filter at Gaussian. Ra is the standard for mechanical surfaces and Rq is the standard for optical surfaces. Second, attach the provided mitutooy to the display unit and the detector.
Turn the unit upside-down, push the probe end of the detector in slightly to release it, and then lift the detector up. Here is a picture of our height gage setup:.
Make sure to read chapter 1 “Installation” in the user manual after downloading and before installing “setup. This software tool can generate a simplified inspection certificate based on the calculation result and evaluation profile data of the Surftest. You probably already own everything you need for this. It requires a computer with Microsoft Excel installed sm-210 operation, but usn really handy for capturing data and printing out inspection reports.
Yes that too is wierd.
Mitutoyo :: Software and Updates
Finally, press Esc a few times or just press the blue button to return to the home display. For information on the parameters that have to be set for error-free communication please refer to the user manual.
I agree to receive the Mitutoyo Ysb. Please use a conventional micro USB cable for data transfer. SJ Detector on a Lens. When you are finished measurint roughness, you need to turn off the SJ and retract the probe for safe-keeping.
Make sure to read chapter 1 “Installation” in the user manual after downloading and before installing by executing “setup. This is shown in the following photo. Everything else may be left at the default setting. The following photo shows the detector on a small plastic lens that is held in place by a piece of modeling clay plasticine. Finally, attach the detector unit to the chosen accessory.
Basic setting and function setting. Before downloading this document you’ll need to register below or login. If you just purchased a Mitutoyo SJ surface roughness tester, you may want to know how to set it up for measuring roughness.
If you watch the display as you are lowering the probe, you will note that the red bar across the top changes to blue, assuring you that the probe is in contact with the sample. Select Setting in the same way.
When you reinstall the detector into the display unit, the probe is reasonably well protected.
Measuring Roughness with an SJ-210
N is the number of consecutive measurements that will be used for averaging. To attach any of these accessories to the SJ, first remove the detector from the underside of the display unit.
What industry are you in?